Important Notes on the replacement of New Supplier’s TP Screen on MaxiSys Tablets.
1.Part Number of new TP screen Involved
Note: The above new TP screen is provided by a new supplier, which will be used for TP replacement of the above models.
Need to do first:
1.Update operating system to the latest version first
E.g. Maxisys 906 V03.46.00, Maxisys906BT V03.31.00 , MY908 V03.41.00
If the MaxiSys was out of update period, please use reflashing program (sent via Wetransfer) to update it to the free version.
1). Remove R289 Resistance for MY908
2). Remove R375 Resistance for MY906/MY906BT
3.Remove specific buttons for MY906,MY906BT,MY906TS
1). Remove J11, J12 buttons(as showed below) for MY06BT, MY906TS
2). Remove J11, J12 buttons (as showed below) for MY906
4.Pasting way of TP for MY906,MY906BT,MY906TS
TP plate should be pasted parallel with the magnesium alloy stent.
Autel MaxiSys Tablet tech support is available on www.auteleshop.com